|Published (Last):||27 August 2006|
|PDF File Size:||19.40 Mb|
|ePub File Size:||10.11 Mb|
|Price:||Free* [*Free Regsitration Required]|
Quick Links. Download this manual. Table of Contents. Compact insulation resistance tester with 3-range 24 pages. Safe Testing Procedure of removing cover Page 3: Safe Testing 1. This instrument must only be used by a competent and trained person and operated in strict accordance with the instructions. KYORITSU will not accept liability for any damage or injury caused by misuse or non-compliance with the instructions or with the safety procedures. Kyoritsu is by no means liable for any damage resulting from the instrument in contradiction to this cautionary note.
The use of other accessories is prohibited as they are unlikely to have the correct safety features. IV, and called measurement categories. The cover can be detached and put on the back side of the main body during measurement. LCD Page 9: Test Lead Barrier : provides protection against electrical shock and ensuring the minimum required air and creepage distances.
Page Applied Standards 3. Auto data hold The LCD reading is automatically frozen for 3 seconds after measurement. It works on basis of current difference between phase currents flowing to different loads and returning current flowing through neutral conductor for a single phase installation. Page 14 This tripping time values are concerning the RCD's correctly mounted following the manufacturer's specifications. Page 15 In order to check the leakage or fault currents there are specific leakage clamp meters as Model , Model , Model F.
Page Test Procedure Do not use this instrument to check the correct wiring of the power supply. Kyoritsu will not be held liable for any accident that may result from incorrect wiring of the power supply line. Page Testing 6. Especially, be careful not to connect it to phase- phase. Please remember to give all the information possible concerning the nature of the fault, as this will mean that the instrument will be serviced and returned to you more quickly.
By hanging the instrument round the neck, both hands will be left free for testing. Pass the strap belt down through the side panel of the main body from the top, and up through the slots of the probe case from the bottom.
Print page 1 Print document 24 pages. Cancel Delete. Sign In OR. Don't have an account? Sign up! Restore password. Upload from disk. Upload from URL.
Kyoritsu 5406A RCD Tester, 10/20/30/200/300/500mA
KYORITSU 5406A Instruction Manual
Kyoritsu 5406A RCD Tester